Deep ultraviolet fluorescence microscopy of three-dimensional structures in the mouse brain

Three-dimensional (3D) imaging at cellular resolution improves our understanding of the brain architecture and is crucial for structural and functional integration as well as for the understanding of normal and pathological conditions in the brain. We developed a wide-field fluorescent microscope for 3D imaging of the brain structures using deep ultraviolet (DUV) light. This microscope allowed fluorescence imaging with optical sectioning due to the large absorption at the surface of the tissue and hence low tissue penetration of DUV light. Multiple channels of fluorophore signals were detected using single or a combination of dyes emitting fluorescence in the visible range of spectrum upon DUV excitation. Combination of this DUV microscope with microcontroller-based motorized stage enabled wide-field imaging of a coronal section of the cerebral hemisphere in mouse for deciphering cytoarchitecture of each substructure in detail. We extended this by integrating vibrating microtome which allowed serial block-face imaging of the brain structure such as the habenula in mouse. Acquired images were with resolution high enough for quantification of the cell numbers and density in the mouse habenula. Upon block-face imaging of the tissues covering entire extent of the cerebral hemisphere of the mouse brain, acquired data were registered and segmented for quantification of cell number in each brain regions. Results in the current analysis indicated that this novel microscope could be a convenient tool for large-scale 3D analysis of the brain in mice.

www.nature.com/scientificreports/ In this paper, we aimed to obtain non-contiguous planar information with significant interval along z-axis to obtain interspersed but three-dimensional distribution of the cells, represented as cell number and density in the specific subregions in the tissue block. This is especially effective in analysing the nervous tissue in which the functionally heterogeneous subregions are distributed in space. We present a simple optical setup for block-face imaging combined with DUV illumination. This method was compatible with 3D imaging of the mouse brain towards quantitative analysis at cellular scale resolution. It is expected that this simplified microscope would be established as a useful tool for 2D/3D microscopy in neuroscientific applications.
Animals. All procedures and experiments were performed in accordance with ARRIVE guidelines (https:// arriv eguid elines. org). The study was approved by the committees of Hiroshima University for the animal experiments (approval number A22-139) and gene recombination experiments (approval number 2022-90). All methods were carried out in accordance with relevant guidelines and regulations. Four male C57BL6/J (Japan SLC, Inc., Hamamatsu, Japan) and two male heterozygous Emx1-Cre mice 13 were used for 2D and 3D imaging, respectively. Three to nine-month -old mice were group-housed under specific pathogen-free conditions with free access to food and water. A 12:12 h light-dark cycle of illumination was maintained under the regulation of the temperature and humidity in the range of 18-25 °C and 40-60%, respectively. They were anesthetized by ketamine/xylazine (100 mg/kg, i.p.) and transcardially perfused with 4% paraformaldehyde in 0.1 M phosphate buffered saline (PBS). The brains were post-fixed in the same fixative at 4 °C overnight.
Tissue staining. For the staining of the 2D sections, we incubated the sections in the dye solutions followed by wash in PBS three times. Reagents and concentrations used in the study was as follows: propidium iodide (5 µg/ml, P1304MP, Invitrogen, USA). For whole brain staining with Hoechst 33258, the osmotic shock method was used as described previously 1,14 . Briefly, fixed brains were immersed in staining solution (PBS containing 10 mg/ml Hoechst 33258, 0.1% TritonX-100, and w/v sucrose: Day1-10%; Day2-20%; Day3 to Day7-30%; and Day8-0%) at 55 °C with gentle shaking. For staining of the block-surface of the mouse brain in situ, the bath of the vibrating microtome was filled with propidium iodide solution (5 µg/ml in PBS) 5 . Propidium iodide stained cytoarchitecture visualized both the nucleus and the Nissl bodies 5,15 .
Wide-field DUV microscope setup. Optical design of the DUV fluorescence microscope is similar to that of conventional epifluorescence microscope, but with fewer optical components (Fig. 1A). Side illumination with oblique incidence of light onto the specimen was used (Fig. 1B, 60° angled relative to the vertical plane of the microscope). A multi-spectral detection scheme with spectrally separated fluorophores in RGB channels was adopted using a colour camera (GS3-U3-123S6C-C, FLIR Integrated Imaging Solutions, Inc., OR, USA). The optical sectioning ability of the DUV fluorescence microscope is due to the strong absorption of the DUV light by the molecules and thus the limited depth of penetration of light 10 . Our DUV microscope design further exploited the oblique light illumination for excitation with water immersion objective for increased angle of refraction and lowered depth of penetration of DUV light 11 .
Light emitting diode (LED) at wavelength 280 nm (incident power on the sample as approximately 8.13 mW, Quark Technology, Okayama, Japan) was used as light source. UV grade fused silica biconvex lens (f = 35 mm) was used to converge the DUV light and limit the area of the illumination. The imaging was carried out using the 10X objective lens (UMPLFLN 10XW, NA = 0.3, Olympus Corporation, Tokyo, Japan). The field of view with the 10X objective was approximately 1.3 mm × 1.0 mm. Image acquisition was performed at default camera settings for exposure time of 30 ms.
For wide-field imaging, a custom-designed motorized stage controlled by Arduino Mega 2560 microcontroller (Arduino LLC, Somerville, MA, USA) was used as a modified version of the open-source based stage controller developed previously 16 (Fig. 1C). The XY stage translates the microscope head to obtain multiple images as mosaics to cover the entire mouse brain section in the chosen anatomical plane. A flexible shaft was connected to 500 µm/revolution micrometer actuated linear translator (X and Z axes: 25 mm translation stage PT1/M; Y axis: 50 mm translation stage XR50P/M, Thorlabs, Japan). Fifty mm translation stage was used for the Y-axis to allow for the complete evacuation of the imaging head for tissue slicing after each block-face imaging. www.nature.com/scientificreports/ Custom-fabricated couplers were used to connect the shaft to the stepper motors (PK243M-01B; Oriental Motors, Japan) and the micrometers of the translation stage. For the stepper motor step size of 0.9°, a minimum resolution of 1.25 µm was obtained with the translators. The master control was carried out using the customwritten software using Python3 which communicates with the Arduino through serial communication (Fig. 1C). The number of X and Y translations required to cover the entire tissue section or tissue section of interest was calculated based on the information provided in mouse brain reference atlas in stereotaxic coordinates 17 with a 20% overlap between the tiles. The parameters corresponding to each of the imaging session was described using YAML files 18,19 .
The widefield imaging capability was extended to carry out whole brain imaging through automated serial block-face imaging (Fig. 1D). This was done by integrating the microscope with a vibrating microtome (Compresstome VF-700-OZ-120, Precisionary Instruments, Greenville, NC, USA) which performs serial sectioning followed by block-face imaging (Fig. 1B). Microtome was controlled using a relay switch (Y14H-1C-5DS; Hsin Da Precision Co. Ltd, Taiwan) through Arduino Mega 2560 to automatically start and stop the sectioning of the tissue after completion of the wide-field data of the previous tissue section (Fig. 1C). Since the acquired images reflect the signal originating from the limited depth (dark blue in Fig. 1E) which is much smaller than the sectioning thickness by vibrating microtome (light blue in Fig. 1E), we obtained non-contiguous planar information with significant interval along z-axis.
Axial profiling of the microscope setup using fluorescent microbead. Four µm microbeads (Tet-raSpeck, Invitrogen, USA) were used for profiling the DUV microscopy setup with 10X water immersion objective (NA = 0.3, Olympus, Japan). The XYZ profiles were obtained as an average of images for 10 microbeads embedded in 2% agarose. Gaussian fit and full width at half maximum were obtained using custom python scripts. www.nature.com/scientificreports/ Serial block-face imaging with DUV. 3D imaging was carried out with the sample placed in the sample holder of the vibrating microtome. The tissue sample was glued onto the sample holder and was then embedded in 2.0% agarose (low EEO Agarose, A0576, Sigma-Aldrich) inside the sample holder. To allow faster setting of the agarose, the bath surrounding the sample holder was filled with chilled PBS solution kept at 4 °C. Then, serial slicing was carried out to expose the tissue surface for imaging. For staining of the brain samples in the microtome bath, PBS is replaced with the respective dye solution as the immersion medium of the objective in the bath. For staining of the block-surface in situ, propidium iodide solution (5 µg/ml) was filled in the bath of the vibrating microtome. Upon exposing the block-face surface of the tissue section of interest to start the image acquisition, images per field of view (FOV) were acquired as per the averaging and Z-stack values input for all the FOVs across the entire wide-field imaging. The XYZ stages for positioning the imaging unit were brought back to the original location after completion of an entire wide-field image acquisition of the exposed tissue section.
To allow the microtome blade to slice off the next section, the imaging head was translated (Y axis) 50 mm away from the original position. The microtome then sectioned the exposed block-face at the desired thickness of 50 µm (imaging of the entire mouse habenula) or 100 µm (imaging of the whole mouse brain) followed by returning of the blade back to the original position. The imaging head was then moved back to the original position and the whole process was repeated till the imaging of the entire block of tissue sample was carried out over serial sections of desired thickness.
Quantification of the optical sectioning thickness. Nissl-stained fluorescence signals using propidium iodide was used for isolating each cells using the co-registered DUV image and confocal z-stacks. Z-stacks were obtained using confocal microscope at depth separation of 2 µm using 10X and 20X water immersion objective lenses (UMPLFLN 10XW, NA = 0.3 and UMPLFLN 20XW, NA = 0.5, Olympus Corporation, Tokyo, Japan). Same specimen was imaged using DUV microscope and co-registered images were obtained. 1345 cells were segmented using a custom developed ImageJ macro. Confocal z-stacks were used to locate the depth of the individual cells and the optical sectioning thickness was assessed by sorting the cells in DUV images based on the depth information. DUV signal attenuation with depth was modelled based on the Beer's law and the 1/e depth of the exponential fit was used to obtain the estimate of the optical sectioning thickness 11 . Statistical measure by averaging over a large number of cells was used for this analysis as varying signal intensities across and within cells for Nissl bodies were observed. Image similarity of the DUV image with the different images of the confocal z-stack at various depth was obtained using Pearson correlation coefficient to corroborate the estimated value. Data analysis. The DUV microscopy images were denoted as RGB for colour images comprising of red, green and blue channels, respectively. Unless otherwise specified, signals in red channel were shown in greyscale intensity levels. The image processing steps included colour correction and flat-field correction. Extended depth of focus imaging was optionally carried out if multiple z-stack images were acquired for each FOV. Widefield mosaic was obtained using Microsoft Image Composite Editor for each of the brain sections. ImageJ was used for further data analysis as described below (https:// imagej. nih. gov/ ij/) 20 .
For 3D quantitative analysis of the cell numbers in the mouse habenula, cells labelled with propidium iodide were counted using Cell Counter plugin of ImageJ for 10 serial sections taken with uniform spacing of 150 µm across the serial sections acquired for unilateral habenula with referring to the anatomical landmarks such as the third ventricle and stria medullaris.
For prototypical whole brain analysis, we consecutively imaged 117 block-face images with interval of 100 µm, spanning from the olfactory bulb to the cerebellum. Eight-bit grayscale images obtained were then converted to 16-bit images for automatic cell counting. Semi-automatic alignment of images to Allen brain reference template [Allen Mouse Common Coordinate Framework (CCF v3)] 21 followed by segmentation and automated cell counting (AP = − 0.92 mm from Bregma) were done using WholeBrain pipeline software 22,23 . The percentage of cell counts was calculated for each brain region and represented as sunburst graph.
Statistical analyses. Statistical analyses were performed using the statistical package R (R Development Core Team 2011), jamovi 1.2 (The jamovi project 2020, retrieved from https:// www. jamovi. org). Binary comparisons for two groups were assessed using two-tailed paired t-tests, and comparisons between three or more groups were performed using Welch's one-way ANOVA followed by Tukey's honestly significant difference (HSD) test for multiple comparisons. Distribution of the signal intensity at different anterio-posterior level of the mouse brain was examined by Shapiro-Wilk test of normality. Statistical significance was defined as p < 0.05.

Results
We first examined the axial resolution of the DUV microscope using 3D profiling of the fluorescent microbeads. Full widths at half maximum for X, Y, and Z axes (4.19, 4.24, and 32.5 µm, respectively) were determined from the virtual slices along XY and YZ planes ( Fig. 2A). This analysis confirmed the 3D resolution high enough for imaging single cells.
Next, we examined compatibility of the synthetic dye and fluorescent probe used for immunohistochemistry with DUV. Spectrophotometry revealed that synthetic dyes frequently used in Hoechst 33258 (nucleus) and propidium iodide for Nissl staining (nucleus and Nissl substance in the perikaryal area) in neuroscience 20 had absorption peaks in DUV range (peaked at 266 and 289 nm for Hoechst 33258 and propidium iodide, respectively) with emitting fluorescence in green and red spectrum (peaked at 496 and 659 nm as shown by blue and red in Fig. 2B, respectively). We also examined donkey anti-rabbit IgG conjugated with Alexa Fluor 594 www.nature.com/scientificreports/ conventionally used in immunohistochemistry and found its absorption and fluorescence peaks at 278 nm and 613 nm, respectively (orange in Fig. 2B). These results suggested that fluorescent probes conventionally used in neuroscience could be, at least in part, applicable to DUV imaging. We applied those probes to the mouse brain section to image the wide field of neural structure covering a hemisphere at cellular resolution by DUV microscope. Propidium iodide excited by DUV emitted signal localized in the cell bodies predominantly in grey matter over white matter on a coronal section of the mouse brain (Fig. 2C). Images exhibited Nissl-like pattern with intense signal in the perikaryal region of the neuronal cell bodies showing six-layered cytoarchitecture in the cerebral cortex (Fig. 2D). Furthermore, this wide field imaging with DUV revealed distribution of the neurons with cell-type specific marker such as tyrosine hydroxylase (TH), a protein marker for dopaminergic neurons in the midbrain. Indeed, immunohistochemistry using anti-TH antibody visualized with Alexa Fluor 594 showed cell bodies were distributed specifically in the substantia nigra pars compacta and ventral tegmental area in the midbrain (Fig. 2E). These images were obtained with high resolution allowing to identify detailed localization of the immunoreactivity in the perikaryal regions, cytosol and proximal part of neuropils (Fig. 2G). Upon imaging of the same sections with conventional epifluorescence microscope, we found images with less contrast and low signal-to-noise ratio probably due to the emitted light from the optical plane out of focus (Fig. 2F,H). Upon combining this with Hoechst 33258 nuclear staining for multi-colour labelling, we observed specific distribution of the Alexa Fluor 594 and Hoechst 33258 in the cytosol (magenta in Supplementary Fig. 1) and nucleus (cyan in Supplementary Fig. 1), respectively. These results indicated the wide-field DUV microscope which we developed in the current study were useful tool to examined neural structures labelled with fluorescent probes for nuclear and Nissl staining and immunohistochemistry at cellular resolution.
Since thick tissue block is likely to have more significant light scattering than cut section as examined above, we addressed applicability of wide-field DUV microscope to the block-face imaging of the mouse brain tissue (Fig. 3). Vibrating microtome exposed the surface of the secondary motor cortex imaged with laser scanning confocal (Fig. 3A,B), DUV (Fig. 3C,D) and epifluorescent microscope (Fig. 3E,F). This was supported by intensity profiling of signals on these cells showing significant difference between three types of microscopes (Fig. 3G, one way ANOVA, F (2,40.8) = 8.55, p < 0.001; Tukey's HSD post hoc test, p < 0.01 for DUV vs epifluoresence). www.nature.com/scientificreports/ To quantitatively assess the optical sectioning thickness, we performed a novel experiment using the method which models the DUV intensity attenuation with depth based on Beer's law and defines the 1/e depth of the intensity attenuation as the estimated optical sectioning thickness as previously described 11 . The results showed that the optical sectioning thickness was estimated to be 20 µm based on the 1/e depth of the DUV intensity of the cells sorted based on depth information from the confocal z-stacks (Fig. 4A,B). Similarity between DUV image and confocal image taken at each depth was assessed with different optical zoom and numerical aperture (Fig. 4C). Quantitative analysis revealed that correlation coefficient between DUV image and a confocal optical section deeper than 20 µm from the surface reduced in line with results of the above analysis (Fig. 4D). This confirms the block-surface optical sectioning capability of the DUV microscope.
We first addressed applicability of our methods to the prototypical case with limited area and promising signal recognition by manual cell counting and registration to the structure. To obtain 3D information from the specific brain structure, we applied the block-face DUV imaging with serial sectioning to the mouse habenula, a longitudinal structure distributing along the rostro-caudal axis (Fig. 5A). Block-surface imaging with in situ Nissl-like staining with propidium iodide exhibited the medial (MHb, yellow in Fig. 5B) and lateral habenula (LHb, red in Fig. 5B). Acquired images were at cellular resolution high enough to allow quantification of the number of neuronal cell bodies in a series of sections (Fig. 5C). Analysis revealed greater cell number and density in MHb than LHb with statistical contrast (Fig. 5D, two-tailed paired t-test, t test, t (9) = 6.75, p < 0.001 for cell number and t (9) = 6.75, p < 0.001 for cell density). Although cell quantification and studying the substructure heterogeneity in habenula is a very important topic, there are few previous studies that quantified the density of cells/neurons in the lateral and medial habenula in the mice. Although a previous study reported a significant difference of cell density between medial and lateral habenula in rats, few studies addressed it in mice 24 .
To extend the results in the above prototypical analysis, we set out more comprehensive analysis covering the whole mouse brain followed by cell counting fully automated by WholeBrain pipeline 22 .We acquired the image datasets from the whole mouse brain to analyse the histological data with annotation of detailed brain structures stained with propidium iodide (Fig. 6A). As in the prototypical analysis examining the habenula, high quality data were obtained from the brain structures ( Supplementary Fig. 2) ranging from the olfactory bulb (Fig. 6B) through the cerebral cortex (Fig. 6C) and the habenula (Fig. 6D) to the cerebellum (Fig. 6E). Within a single plane in the middle of entire anterio-posterior extent, five different field of views with fixed depth of focus showed signal-to-noise ratio high enough to discern each of the cellular nucleus ( Supplementary Fig. 3). The staining of propidium iodide was uniform along the anterior-posterior axis of the mouse brain, since we observed mean signal intensity sampled from six rectangular fields at each level followed Gaussian distribution (Shapiro-Wilk test of normality, W = 0.965, p = 0.001) and did not show statistically significant difference   Fig. 4). Cell count data was mapped onto coronal planes (Fig. 6F) and 3D annotation space (Fig. 6G) of the Allen brain atlas to segment the anatomical subregions. This allowed the quantification of the number of cells in a hierarchy of brain regions as represented by sunburst plot (Fig. 6H,I).

Discussion
Microscope in the current study is different from the conventional microscope such as confocal microscope with optical sectioning ability in that it images a surface of the block-face to obtain the structural information of the tissue at a given depth. By consecutive sectioning using vibrating microtome to expose new and deeper block-face for imaging, three-dimensional information of the tissue is thus obtained. Block-face imaging using epifluorescent microscope can result in blurred image due to the emitted light coming from the plane in and out of focus 25 . To address this, we adopted DUV light source which attenuates in the plane deeper than the superficial part of the tissue thus providing optical sectioning capability for block-face imaging. We combined DUV with vibrating microtome which enables the sectioning of the fixed samples without intensive preparation and embedding of the samples in advance. We aimed to obtain non-contiguous planar information with significant interval along z-axis (e.g., 50 µm) to obtain interspersed but three-dimensional distribution of the cells represented as cell number and density in the specific subregions in the tissue block. This is especially effective in analysing the nervous tissue in which the functionally heterogeneous subregions are distributed in space.
Similar approach was adopted by a previous article in which the DUV was combined with rotary microtome to image the block-face of the samples embedded in paraffin wax following intensive tissue processing such as dehydration, clearing and embedding 26 .Although it allows imaging with finer z-resolution due to the capability to cut thinner sections (by 1 µm) than the current study, we believe that microscope in the current study has advantage in obtaining interspersed information covering major brain regions distributed in the forebrain quickly. Registration of the cellular information to the reference brain atlas using pipeline software further supports its use in processing a large number of samples as frequently found in group comparison with different genotypes and drug treatments in rodent studies.  www.nature.com/scientificreports/ The underlying principles of DUV microscope are the same as that of the recently developed microscopy with MUSE 10,11 . MUSE has been proposed as a rapid tool for obtaining histology like images for pathology labs. In this study, we show the possible extension of the utility of MUSE/DUV microscope as an alternate for conventional epifluorescence microscope used in neuroscience research for block-face imaging. DUV microscope presented here has a simple optical design that could be convenient alternative to the conventional microscope in research labs for both 2D and 3D microscopy 12 .
From a design perspective, oblique illumination is utilized for DUV microscope unlike the objective lens system for excitation and emission light collection in the conventional fluorescence microscope design. Objective lens with higher numerical aperture has benefits for higher magnification and collection of more light from the samples. But high numerical aperture lens has shorter working distance which can limit the angle of incident light illumination from a design perspective.
Estimated optical sectioning thickness in the current study is slightly higher than the one measured in DUV microscope reported previously 11 for similar microscope configuration. Since these two studies have difference regarding the samples (fixed mouse brain and unfixed human breast tissue) and staining (exclusive labelling of cell nuclei and labelling of both nuclei and perikaryal Nissl substance), it is likely that differential distribution of the subcellular signal and DUV light scattering in the tissue contributes to the estimated optical sectioning thickness. Further quantitative assessment under different sample conditions would address these points in the future.
It is essential to create the list of the dyes with high fluorescence yield for excitation using DUV spectrum. Previous studies reported that DUV light excited a range of fluorophores including synthetic dyes such as Hoechst 33258, propidium iodide 10 and quantum dots 27 . In the current study, we showed that immunohistochemistry with Alexa Fluor dye conjugated with antibody was compatible with DUV imaging. In addition to the antibody, Alexa Fluor dyes exhibited high photostability and brightness and were applied to labelling of the anterograde and retrograde tracers widely used in the analysis of neural circuits 28 . Further analysis of the compatibility of those reagents with DUV would enhance the applicability of DUV imaging in neuroscience.
For the application of the block-face imaging to the DUV imaging, whole brain staining with antibody must be essential in advance. Since whole brain staining with antibody were reported with intensive tissue processing for clearing to facilitate antibody penetration, reduced tissue integrity would be a significant concern upon sectioning in block-face imaging 29,30 . Future study is needed to screen the clearing procedure compatible for vibratome sectioning to address this point.   www.nature.com/scientificreports/ Whether or not UV would pose a problem for tissue imaging might be a concern in general. In the current study, we typically used 6-8 mW of LED power for a brief exposure time of 30 ms, which did not seem to possess problems in tissue imaging for the range of dyes which have explored. Indeed, we observed a preserved cytoarchitecture of the mouse brain with frequent DUV imaging as in the conventional microscopy under visible light. DUV microscopy for fluorescent probes used in immunohistochemistry showed rather better signal-tonoise-ratio as compared to the epifluorescent microscopy. In line with this, a previous study on live cell imaging using DUV light have shown that UV phototoxicity is not a concern for live cell imaging to study phenomena such as mitosis and motility for UV exposure for as long as 45 min 31 .

MHb
Illuminating the tissue with DUV also involves the significant presence of autofluorescence coming from the endogenous UV excitable biomolecules attributed to excitation of the aromatic amino acid, nicotinamide adenine dinucleotide hydrogen (NADH) and collagen 32 . Although these molecules are excitable in the range from 220 to 295 nm, their emissions are limited from 280 to 350 nm and outside of the visible spectrum. Thus, it is unlikely that they affect the emitted fluorescence of exogenous fluorophores used in the current study. The fluorescence of the endogenous biomolecules with an improved contrast for brain imaging in cancer imaging was recently shown 33 . In the brain samples, white matter autofluorescence with DUV excitation is significant 34 . Although we did not spectrally separate the autofluorescence in the white matter in the current study, extraction of the signal reflecting the autofluorescence from the specific brain structure such as white matter might be helpful to demarcate the anatomical landmark to gain better insight into the brain structure.   www.nature.com/scientificreports/ Within the diffraction limited regime, use of the fluorophores with shorter wavelength of the DUV spectrum has the added advantage of better spatial resolution, while capability of microtome in cutting thin section will limit resolution along z-axis in block-face imaging. For example, rotating and freezing microtome commonly used in the histological analysis allows finer sectioning than vibrating microtome, although they require additional processing of the tissues in advance of sectioning such as dehydration, clearing and embedding 26 . In contrast, vibrating microtome used in this study allows sectioning of the fixed tissue without extensive tissue embedding, which is beneficial to preserve more antigenicity in the tissue to make it compatible with immunohistochemistry. Taking the cellular resolution analysis of the mouse brain in the current study into consideration, sectioning tomography with a combination of DUV and vibrating microtome could be a useful solution to analyse the structural and neurochemical properties of the brain.

Conclusions
We developed a novel microscope for wide-field imaging of 3D structure of the mouse brain using DUV illumination. With integration of the motorised stage, this microscope could cover a coronal section of the entire hemisphere stained with synthetic dyes and immunohistochemistry for protein localization. Imaging of 3D structure in mouse brain was done upon block-face imaging with serial sectioning by vibrating microtome. We showed prototypical analysis of the cell distribution in the entire mouse brain with reference to the standardized mouse brain atlas. Results in the current analysis indicated that this novel microscope could be a convenient tool for large-scale 3D analysis of the mouse brain.